The Gocap screen displays the insulin value that it measures from the insulin pen.
The values displayed in the app are as accurate as the ones displayed on the Gocap. Gocap is accurate within +/- 1 unit for doses up to 10 units, and +/- 10% of the dose for doses above 10 units, but occasionally there will be outlying data points.
Some possible causes of outliers are:
Keep in mind the doses that Gocap records are only as accurate as each step in the dosing procedure such as the injection technique, the insulin pen, and the use of the Gocap itself.
For additional information, contact the Support line (855) 310-3377 or email email@example.com
The Gocap website requires cookies to be enabled in order to power the shopping cart. Please enable cookies in your browser and refresh the page.